Monitor - ISSN 1472-0221
The Newsletter for Data Acquisition and Control
Issue 263 September 2020
Hello and thank you for subscribing. I hope you find the newsletter useful.Comments to [email protected].
You can download a pdf copy of this newsletter here.
Carbon dioxide is a greenhouse gas which contributes to global warming. Its concentration is rising in atmosphere and directly increasing extreme weather events. The gas also dissolves into the ocean, causing ocean acidification.
To mitigate the effects of excess CO2, it can be trapped within reservoirs of rock. One way to do this is to dissolve CO2 into the brine which is within pores of rock under the sea. However, there are concerns on the effect of the stored CO2 on the reservoir rock, given the need to ensure the long term integrity of the storage reservoir. Injecting CO2 stresses the rock and affects its fatigue behaviour.
Engineers in Edinburgh and Wolverhampton are investigating how the behaviour of CO2 in rock is controlled by temperature, pressure and saturation conditions. As part of their investigations they are using Windmill software and the Microlink 751 multi-function data acquisition unit.
The experimental set up held the rock sample, brine and CO2 at 35 oC during fatigue testing. A pressure transducer measured and logged the pressure on the sample. The Microlink 751 recorded displacement data from LVDTs and sent the readings to Windmill software running on a PC to display and save the data. The researchers set the Microlink at 15 bits per channel with each channel measuring +/- 10 V dc. They chose a resolution of 0.8 mV giving a precision of +/- 0.05%.
Axial and lateral displacements were recorded per second during the entire fatigue and Windmill displayed the data on the screen. The readings were processed to yield the axial and lateral strain, volumetric strain, fatigue rates, fatigue duration, bulk modulus and compressibility.
The Microlink can measure temperature, strain, pressure, voltage and current through 16 analogue input channels. It can switch up to 32 digital outputs and monitor up to 32 digital inputs. It can also count events with up to 8 counters. You can connect eight Microlinks to one PC giving 128 analogue inputs and 256 digital inputs and outputs. Use Windmill software to choose from four analogue input ranges or automatic ranging to let the software match the input signal as closely as possible. You can also use Windmill to select the resolution from 7 options: choose high throughput or high resolution. Read more
Ameh Peter et al, Static fatigue of saline rocks under different CO2 phase conditions, Journal of Petroleum Science and Engineering, Volume 195, December 2020, 107940 Advance Article
The Microlink 751 multi-function data acquisition unit
Hugh Baker et al, Rising CO2 may increase dangerous weather extremes, whatever happens to global temperatures, University of Oxford
We do have two Microlink 751 with each having a 590-8A connected to it and Windmill 7. We would like to log signals from 0 - 20 mA. Could we have an explanation how to do the setup for 6 channels of 0 - 20 mA?
Connect the 0-20 mA signal to the + and - connections of one channel in the in the 590 box. For 6 inputs this will be for channels 0 to 5. You should also connect the - inputs to ground.
To the Windmill software the inputs will be like normal analogue voltage inputs, because the 8 channels of current inputs have 100 ohm resistors, which make 20 mA into a 2V signal, and 0 mA into 0 V. Use the SetupIML program to name the engineering units.
For more on the Microlink 751 see https://www.windmillsoft.com/daqshop/daq-usb.html.
Researchers develop smallest particle sensor in the world
Smartphones, smart watches or fitness wristbands can for the first time measure the quality of the ambient air in real time and sound the alarm in the event of increased fine dust values.
Sensitive strain sensor that can detect the touch of a feather
The sensor can stretch up to 80 times higher strain than strain gauges currently on the market and show resistance changes 100 times higher than the most sensitive materials in research development.
Source: University of Sussex
Velcro-like food sensor detects spoilage and contamination
Color-changing array of silk microneedles could help protect consumers and avoid food waste
Source: National Science Foundation
Environmental Sensing and Monitoring Technologies to grow by 6 billion
Emerging growth areas include such as large-scale monitoring networks.
Source: BCC Research
Sound waves show warming oceans
Seismic monitoring equipment, and historic seismic data, determine how much the temperature of the earth's oceans has changed and continues changing, even at depths that are normally out of the reach of conventional tools.
Source: SCUBA News
* Copyright Windmill Software Ltd
* Reprinting permitted with this notice included
* For more articles see https://www.windmill.co.uk/
We are happy for you to copy and distribute this
newsletter, and use extracts from it on your own web
site or other publication, providing the above notice
is included and a link back to our website is in place.
For previous issues by subject see https://www.windmill.co.uk/monitorindex.html
SUBSCRIBING OR CANCELLING SUBSCRIPTION Visit https://www.windmill.co.uk/newsletter.html and add or remove your e-mail address.
Windmill Software Ltd, PO Box 58, North District Office,
Manchester, M8 8QR, UK
Telephone: +44 (0)161 833 2782
Facsimile: +44 (0)161 833 2190
E-mail: [email protected]